1. Spin only as fast as needed for your experiment. Just because your car can go 200 km/h does not mean that it should be driven at 200 km/h. Likewise, just because your probe is rated to spin samples at 15 kHz does not mean that all samples should be spun at 15 kHz.
2. Check the integrity of the rotor and the cap before use. Damaged rotors are weakened and should NEVER be used. Damaged caps can cause instability which may lead to a rotor crash.
3. Make sure the cap fits snugly on the rotor. A cap that lifts or comes off the rotor while spinning will cause a rotor crash.
4. Be aware of sample heating due to spinning. The rotor and sample heat up during spinning due to friction. If the temperature increases such that your sample melts or emits a gas, the rotor may become unbalanced or the cap may be forced off causing a rotor crash.
5. Make sure the rotor is marked properly so the speed can be monitored and regulated. Failure to do this may result in the spin counter receiving a bogus signal and it is possible that the rotor may spin faster than its rated speed causing a crash.
6. Pack your sample evenly to ensure that the rotor is properly balanced during MAS. Rotors that do not spin smoothly and stably should be repacked until they do.
7. Start and stop the rotor gradually to ensure stability while speeding up or slowing down.
(Photo courtesy of Paul Morris of Morris Instruments Inc., manufacturer of very useful probe tuning devices and provider of magnetic resonance products and services.)